This service note describes how to perform the installation of the I/O board for NM Plus and PG Plus hydrogen generators.
The VICI DBS electrolysis cell is a Proton Exchange Membrane (PEM) product. One of the characteristic of this technology is, that the cell, or, better, the membrane in the cell, must stay wet at any time. In case of complete dry out of the inner part of the cell, the membrane can be damaged and perform less Hydrogen afterwards. In most cases, this damage is not reversible and the cell need to be serviced.
Hydrogen generators and cells contains water. Cells can be damaged if temperature is below 0 °C.
This service note explains how to remove the transport tubing from hydrogen cell. Be careful not to damage the water connector. Leakage could occur.
Several improvements have been implemented. All new generators are equipped with water pump and flow sensor. Gas Liquid separator and water pump are placed next to each other on the bottom and more easy accessible to maintain.
The water drain line is separated from the water circulation. Only 1 auto-refill kit is necessary per cascading system.
This service note informs about a change on the cassis. From S/N FNMX161215053 the top cover is split in two parts for easier service access.
This Manual describes how to use the I/O board. How to install the I/O board see Service note 2017/001.
This Manual describes how to perform a routine maintenance on external water tank and circulation pump on FID station / FID tower and Rack generators.
Virtual COM port (VCP) drivers cause the USB device to appear as an additional COM port available to the PC. Application software can access the USB device in the same way as it would access a standard COM port.
This Service note describes how to install the driver.
When the generator is started, the internal pressure of 11 bar needs to be built up. This pressure is monitored by a sensor on the dryer module. If the rate of increase is not within the tolerance as set in the firmware, the H2 production will be stopped and the error message “Low int. pressure” is shown.
There are 4 major sources for this problem:
defective Gas Liquid Separator
leak in the cell
defective dryer module
major leak in tubing & fittings
This service note describes how to reset the “catalyst replacement warning” LED on NGC and NGT boards.
The catalytic converter (catcon) has a limited lifespan and therefore needs replacement after 3 years of operation (please see the operation manual). As a reminder to the operator, the generator will display a warning message “Catalyst life alarm”, when the life of the catcon is exceeded. The red external “check system” LED begins to flash. From this time on, the operator has 60 days to replace the catcon. When this time expires, the control board disables the system.
From the beginning of 2013 we have changed the mainboard for Air Generators.
This caused a non-retrofit compatibility for GC1500 and GT1500 model below serial numbers.
In the past it had to be defined on the order for which zero air generator model the board is used. To avoid some confusion and make the service easier new part numbers are created and the old one set to obsolete.
The LED on the front shows the status of the instrument. In case of an error the motherboard has a
small single-number display used to indicate an identifying code.